Good Quality Xenon Environmental Solar Radiation Test Chamber - Temperature Humidity Vibration Combined Climatic Test Machine – Hongjin
Good Quality Xenon Environmental Solar Radiation Test Chamber - Temperature Humidity Vibration Combined Climatic Test Machine – Hongjin Detail:
The temperature and vibration three comprehensive test system is a combination of high and low temperature circulation system electromagnetic vibration test system, used to simulate aviation,
Aerospace, automotive, electronics and other products in a more realistic environment, such as; sports vibration impact in high temperature heat resistance, low temperature cold resistance, vibration and other environmental adaptability to products and components and defects in exposed products.
This equipment can perform the following single or combined (integrated) reliability tests: high temperature storage test, low temperature storage test, high and low temperature cycle
Ring test, sinus test, random test, impact test, vibration search and residence test.
Electromagnetic vibration system performance indicators
Shaking table principle
The electromagnetic vibration test system works like a speaker. That is, the energized conductor is moved by the electromagnetic force in the magnetic field.
When the moving coil in the magnetic circuit of the vibrating table passes the alternating current signal, the exciting force is generated, and the vibrating motion is generated in the magnetic circuit.
|Sinusoidal excitation force:||2000kg|
|Random excitation force:||2000kg|
|Frequency range:||5-3000HZ (Taiwan index)|
|Moving coil diameter:||Φ280mm|
|Sports parts quality:||20kg|
|Vibration waveform:||sinusoidal vibration, random vibration, high frequency vibration, shock, resonance search and residence.|
|Product fixture:||An user request for a set of fixtures|
Vertical expansion countertop
1. Table size: 400 * 95mm (round)
2. Countertop weight: 8.0kg
3. Upper limit frequency: 3000Hz
4. Countertop material: aluminum-magnesium alloy
Shaking table body features:
1. Double magnetic circuit structure: Compared with a single magnetic circuit, the magnetic field strength is strong, the volume of the platform is small, and the magnetic leakage is small.
2. No skeleton moving coil: due to the elimination of the moving coil skeleton, the active system is light in weight, the no-load acceleration is large, and the first-order self-vibration frequency of the moving coil is high.
The distortion of the full-band waveform is small, and the dynamic heat dissipation is good.
3. Suspension device: the roller arm suspension system and the linear bearing guide, which makes the vibration table have good guiding performance and good anti-offset capability.
4. Air spring support with small dynamic stiffness and large static stiffness is adopted: the bearing capacity is strong and the low frequency characteristics are good.
5. Trunnion vibration isolation: When the vibration is vertical, the dedicated foundation can be omitted.
6. The size of the vibrating table is about 900mm × 670mm × 960mm
7. The weight of the platform is about 1600kg
Temperature test chamber performance index
1. Temperature range: -70°C to +150°C (arbitrarily set in range)
2. Temperature fluctuation: ±0.3 °C
3. Temperature deviation: ±1.0 °C
4. Heating rate: 1.0 ~ 10 °C / min (slope can be set)
5. Cooling rate: 1 ~ 10 °C / min (slope can be set)
Remarks: The above performance indicators refer to the data tested under the condition that the temperature and humidity are stable for 30 min without load.
Air circulation system
1. Air supply principle: The built-in air conditioning box of the test box is designed on the top of the box, and the stainless steel centrifugal fan is installed on the top of the box, through the fan worm
The shell outlet guide vanes are evenly fed into the working chamber from the top of the tank to meet the requirements for wind speed and temperature uniformity.
Seeking to achieve heating, humidification, cooling, constant temperature control of the studio.
1. Heating method: electric heater heating air circulation mode
2. Heater: heater made of bare spiral nickel-chromium heating wire and insulating ceramic bracket; solid state relay for non-contact zero-crossing
The trigger drive heater works, no contact sparks and noise, and long service life.
3. Temperature measurement: High-precision imported Finnish platinum DIN PT-100Ω temperature measuring sensor.
4. Temperature control: The controller compares the temperature sensor signal of the cavity temperature and the set value, and controls the output through PID+SSR.
The heating power or refrigerant flow is automatically adjusted to achieve a dynamic constant temperature balance of the air temperature inside the equipment box.
1 On-site lecture: The professional after-sales service personnel will train the users on site.
2 On-site guidance: During the execution of the project, our engineers will explain in detail in the actual operation.
3 operation steps, instruct personnel to operate, and answer the corresponding technical problems.
Installation and commissioning method, equipment principle, equipment structure, controller operation, basic function operation, general fault
Diagnosis, positioning and elimination, precautions, routine maintenance.
Product detail pictures:
Related Product Guide:
Assume full responsibility to satisfy all needs of our clients; achieve continual advancements by endorsing the expansion of our purchasers; turn into the final permanent cooperative partner of clientele and maximize the interests of clients for Good Quality Xenon Environmental Solar Radiation Test Chamber - Temperature Humidity Vibration Combined Climatic Test Machine – Hongjin , The product will supply to all over the world, such as: Brasilia , Nepal , Auckland , We adhere to the honest, efficient, practical win-win running mission and people-oriented business philosophy. Excellent quality, reasonable price and customer satisfaction are always pursued! If you are interested in our items, just try to contact us for more details!
By Frances from Morocco - 2017.07.07 13:00
A good manufacturers, we have cooperated twice, good quality and good service attitude.
By Daisy from Hungary - 2018.06.30 17:29